Umbilical Cable Assembly
© 2013-2021 Aer Sampling USA
Meets or exceeds the following methods/laws/requirements:
Section 8.4.2 (Pretest Leak Check), US EPA Method 5
Section 8.4.3 (Leak Checks During Sample Run), US EPA Method 5
Section 8.4.4 (Post-Test Leak Check), US EPA Method 5
Connects various products (pitot tubes on probe sheaths, probe heaters, Type K thermocouples, heater boxes (Hot Box), and umbilical cable adapter) to metering consoles.
Connects with equipment from other manufacturers as well, by using Aer Sampling Compatibility Adapters (Compdapters).
This product connects to the following (non-exhaustive) list of products from Aer Sampling:
- Aer Sampling Meter Consoles
- Aer Sampling Probe Heaters
- Aer Sampling Probe Sheaths
- Aer Sampling MKORTM Heater Boxes (Hot Boxes)
- Aer Sampling Compatibility Adapters (Compdapter)
- K-175 --- Thermocouple Assembly for Double L Connector Kit
- PN-379 --- Umbilical Cable Adapter Assembly
Pitot line (black tubing): 1/4 in. quick connect body; at both ends
Pitot line (white tubing): 1/4 in. quick connect stem; at both ends
Sample line: 1/2 in. quick connect; stem and body
Electrical: 5-contact circular connector; pin and socket
Thermocouple: Type K, standard size, jacks and plugs with labels (PROBE, STACK, FILTER, OVEN, EXIT, AUX)
Weight (Gravitational): ~29 lbs / 13.25 kg
Packing dimensions (Length x Width x Height):
Store this product loosely (i.e. don’t coil it too tightly) to avoid tubing from developing cracks which will cause leaks in time.
All our products come with a one-year warranty against manufacturing defects and one-month technical support for free. Please refer to our Frequently Asked Questions (FAQs) page for more details.
Information presented here is subjected to Aer Sampling Group's terms and conditions.
In case of discrepancy between English version and the non-English version, the English version shall prevail.
Doc ID: AERHQWW-pn-1379-en
Last revised (Excluding prices): 0; 02.10.21
Obsolete Catalog ID: PN-1271, PN-359