© 2013-2021 Aer Sampling USA
Used to heat up temperature of sampled flue gas flowing through probe liner before it reaches filter. Probe liner excluded.
4ft or 6ft
Fits all probe liners from Aer Sampling unless otherwise mentioned.
Purpose and how to use? Relationship to other products?
The probe heater is used as a sort of "pre-heater" in order to make sure that sampled flue gas hits filter at the required temperature. Especially when the nozzle is positioned at traverse/sampling points that require most of the length of the probe to be outside the chimney/stack.
Section 184.108.40.206 (Probe Liner), USEPA Method 5
Section 6.2.3 (Suction Tube - sample probe), ISO 9096:2003(E)
Additional ordering/shipping procedures:
None. Please refer to the FAQs page for shipping and other charges.
Information presented here is subjected to Aer Sampling Group's terms and conditions.
Doc ID: AERHQWW-pn-363-en
Last revised (Excluding prices): 2; 09.15.20
|Catalog ID||Previous (Obsolete) Catalog ID|